for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Published (Last):||5 November 2011|
|PDF File Size:||8.22 Mb|
|ePub File Size:||16.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
System-related Operations Chapter 3 Features and Functions System-Related Operations System-Related Operations This section gives information on topics such as power-down recall, self-test, error conditions, and front-panel display control. Burst Modulation Overview The following is an overview of the steps required to generate a burst-modulated waveform.
To Store The Instrument State Chapter 1 Quick Start To store the instrument state To store the instrument state You can store up to three different instrument states in non-volatile memory. Select from 10 mHz to 15 MHz kHz for triangle and ramp.
Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the triggered sweep mode, the function generator outputs a single sweep each time a trigger is received.
Agilent A Function Generator – Driver – File Exchange – MATLAB Central
Set the carrier frequency between 10 mHz and 15 MHz kHz for triangle and ramp. Turn on power triac – proposed circuit analysis 0.
Be the first to write a review.
There are only two ways to tell somebody thanks: Digital multimeter appears to have measured voltages lower than expected. Message 1 of Contents Chapter 1 Quick Start To prepare the function generator for use 15 If the function generator does not turn on 16 To adjust the carrying handle 18 To set the output frequency 19 To set the output amplitude 20 To set a dc offset voltage 21 To set the duty cycle 22 To output a stored arbitrary waveform You are now following this Submission You will see updates in your activity feed You may receive emails, depending on your notification preferences.
The value or equivalent heating value specifies the ability of the ac signal to deliver power to a resistive load heat. Chapter 1 Quick Start To set a dc offset voltage To set a dc offset voltage At power-on, the function generator outputs a sine wave with a dc offset voltage of 0 volts into a 50 W termination.
But is that same as disabling the manuwl Government prime and Service Office. Block length must be even The function generator represents binary data for 331120a waveforms as a bit integers, which is sent as two bytes DATA: Message 6 of The function generator uses line to send a hold-off signal. By pressing ‘print’ button you will print only current page.
Select a Web Site
Cal security disabled by jumper 33120x calibration security feature has been disabled with a jumper inside the function generator. You can download between 8 and 16, points per waveform. How reliable is it? The GPIB interface is selected when the function generator is shipped from the factory. ALL Delete all user-defined arbitrary waveforms from memory.
I didnot find the option in the manual. You may never have to change some of the parameters discussed here, but they are provided to give you the flexibility you might need. Chapter 3 Features and Functions Output Configuration Instrument State Storage You can store up manaul three different instrument states in non-volatile memory.
Choosing IC with EN signal 2. Sweep Commands see page for more information. Repeat this step as needed. You can issue triggers for bursts and sweeps using internal triggering, single triggering, or external triggering.
PV charger battery circuit 4. Duty cycle is specified as a percentage and represents the amount of time per cycle that the square wave is high.
After one sweep from the start frequency to the stop frequency, the function generator waits for the next trigger while outputting the start frequency. Chapter 5 Error Messages Calibration Errors Calibration Errors The following errors indicate failures that may occur during a calibration.
Page 31 Front-Panel Menu Operation